Information for Healthcare Professionals
Our plan to address the CoaguChek XS PT Test Strip calibration issue
Roche Diagnostics is dedicated to offering quality products that deliver the performance you expect at the point of care. To help you provide the best care to your patients, here is what you should know about the recent recall of CoaguChek XS PT Test strips. To see Roche Diagnostics' latest media release addressing this recall, click here.
Roche Diagnostics, the manufacturer of CoaguChek meters and test strips, recently calibrated the CoaguChek XS PT Test Strips to the most recent International Normalized Ratio (INR) Standard. Since calibrating to this new standard, Roche Diagnostics has been informed of patients experiencing inaccurately high INR test results when testing with the affected test strips. As a result, Roche Diagnostics issued an Urgent Medical Device Correction (UMDC) in September 2018 instructing patients and healthcare practitioners to confirm any CoaguChek test results above an INR of 4.5 with another testing method. This UMDC was distributed to CoaguChek distributors, customers, healthcare professionals and patients via mail.
Roche Diagnostics is now recalling affected CoaguChek XS PT test strips and shipping new lots of CoaguChek XS PT Test Strips that are not affected by the recall and do not require testing with another method. We intend to replace all CoagChek XS PT Test Strips in the market that are affected by the recall with new unaffected test strips as quickly as possible.
Stop using and discard any CoaguChek XS PT Test Strips listed in the table. To see the latest media release indicating lot numbers, click here. Also, please advise your self-testing patients to do the same. Roche Diagnostics has started shipping unaffected test strips to healthcare providers, patient self-testing service providers and distribution partners.
If you have questions about this information or have a product complaint, please call Roche Point of Care Technical Support at 1-800-428-4674 24 hours a day, 365 days a year.
As a healthcare provider, we want to keep you informed with the latest product news and educational information. Please submit this form below to receive ongoing updates.
Roche Video PT INR Testing: Calibration, Correlation, Concordance
This website contains information on products which is targeted to a wide range of audiences and could contain product details or information otherwise not accessible or valid in your country. Please be aware that we do not take any responsibility for accessing such information which may not comply with any legal process, regulation, registration or usage in the country of your origin.